I have a few questions regarding connecting an oscilloscope (using the schematic found here) to CW-Lite. Soldering some pinheaders (or connectors) as needed is not a problem.
Where on the CW-Lite board (PCB) can I connect an external coax trigger signal output from my oscilloscope? Want to let the oscilloscope handle triggering.
Are there any points on the CW-Lite board where I can hook up an oscilloscope with passive probes and use the CW-Lite ADCs as a “diff probe” with the CWCapture program running?
Instead of breaking off the Device Under Test (DUT) board, is it OK to just remove solder bridges SJ4 and SJ5 to use external target? Assuming I can just use pinheaders with jumpers on JP7 and JP6 to enable/disable the internal target as needed then.
BTW: I’m impressed with the design, software and tutorials so far, learning a lot, thanks.